The different types of nano to submicron particles, which are well-dispersed nano particles, coagulated nano particles or settled submicron particles, have been measured by the interactive force apparatus (IFA). The information is very useful for mineral processing treatment. The IFA can measure the particle size distribution in low to high concentration of particles in liquids and non-light transparent liquids. Here, mainly the emulsion and bubble size measurements are described. Emulsion (w/o type) is prepared with oil and water mixture. The nano to micro emulsion particle size is usually measured by light scattering method. The particle size distribution of emulsion of light oil measured by IFA agreed well with the one by the dynamic light scattering (DLS) method. The water size distribution in emulsions of gasoline is difficult for the small reflection index and C heavy oil is difficult for the black color by DLS method. The IFA can measure both emulsions. There are many methods to measure fine air bubbles, however, the finer nano size measurement apparatuses are DLS method and IFA. A “ghost” peak appeared in the large particle micron meter size in DLS method. The mean diameter in IFA is similar with DLS for the nano meter size.